Reading Temperature and Dissolved Oxygen1. Before leaving to sample a lake, check to make sure that there are no air bubbles under the probe membrane of the YSI TemperatureorDissolved Oxygen meter. If there are air bubbles or if it has been several months since changing the membrane (or if the instrument does not calibrate well or the oxygen readings wander), change the membrane as explained in the manual. Note: We have always used the Standard membranes. If adding water to new membrane fluid bottle (KCl), make sure to add Milli-Q water and not the CFL distilled water.2. Be sure to always store the probe in 100percent humidity surrounded by a wet sponge or paper towel.3. Turn on the temperatureordissolved oxygen meter at least 30 minutes before using it. It is best to turn it on before leaving to sample a lake as it uses up batteries slowly.4. Calibrate the meter using the chart on the back of the instrument (adjusted to the Madison altitude - 97percent oxygen saturation). Leave the plastic cap on the probe (at 100percent humidity). The temperature should not be changing during the calibration. Zero the instrument. When the temperature equilibrates, adjust the oxygen to correspond to the chart. After calibrating the instrument, switch the knob to percent oxygen saturation to make sure it is close to 97percent.5. Take readings at 1 meter intervals making sure to gently jiggle the cord when taking the oxygen readings (to avoid oxygen depletion). Jiggling the cord is not necessary if using a cable with a stirrer. Take half meter readings in the metalimnion (when temperature andoror oxygen readings exhibit a greater change with depth). A change of temperature greater than 1degreeC warrants half-meter intervals.6. Record the bottom depth using the markings on the temp.oroxygen meter cord and take a temperature and dissolved oxygen reading with the probe lying on the lake bottom. Dont forget to jiggle the probe to remove any sediment.7. If any readings seem suspicious, check them again when bringing the probe back up to the surface. You can also double check the calibration after bringing the probe out of the water (and putting the cap back on). Light (PAR) extinction coefficient is calculated by linearly regressing ln (FRLIGHT (z)) on depth z where the intercept is not constrained. FRLIGHT(z) = LIGHT(z) or DECK(z) where LIGHT(z) is light measured at depth z and DECK(z) is light measured on deck (above water) at the same time.For open water light profiles, the surface light measurement (depth z = 0) is excluded from the regression.For winter light profiles taken beneath the ice, the first light data are taken at the bottom of the ice cover and are included in the regression. The depth of uppermost light value is equal to the depth of the ice adjusted by the water level in the sample hole, i.e., the depth below the surface of the water. The water level can be at, above or below the surface of the ice. If the water level was not recorded, it is assumed to be 0.0 and the calculated light extinction coefficient is flagged. If ice thickness was not recorded, a light extinction coefficient is not calculated.For light data collected prior to March, 2007, light values less than 3.0 (micromolesPerMeterSquaredPerSec) are excluded. For light data collected starting in March 2007, light values less than 1.0 (micromolesPerMeterSquaredPerSec) are excluded. Except for bog lakes before August 1989, a light extinction coefficient is not calculated if there are less than three FRLIGHT values to be regressed. For bog lakes before August 1989, a light extinction coefficient is calculated if there are least two FRLIGHT values to be regressed. In these cases, the light extinction coefficient is flagged as non-standard.FRLIGHT values should be monotonically decreasing with depth. For light profiles where this is not true, a light extinction coefficient is not calculated.For samples for which light data at depth are present, but the corresponding deck light are missing, a light extinction coefficient is calculated by regressing ln (LIGHT (z)) on depth z. Note that if actual deck light had remained constant during the recording of the light profile, the resulting light extinction coefficient is the same as from regressing ln(FRLIGHT(z)). In these cases, the light extinction coefficient is flagged as non-standard.